- SOFIA Overview
- Proposing & Observing
- Meetings and Events
Download the PDF Version
Return to the Table of Contents for this section at any time by selecting Return to Table of Contents . Users may also navigate through the entire Observer's Handbook by using the complete Table of Contents menu to the right.
6.1.1 Instrument Overview
The Focal Plane Imager (FPI) is the standard tracking camera for the SOFIA telescope. Since an imager upgrade in 2013, the FPI uses a science grade CCD sensor and is referred to as FPI+ (FPI_PLUS). Since the FPI+ is a subsystem of the SOFIA tracking system, it is permanently installed on the telescope. Therefore, it can be operated on every observing flight, either stand-alone or in parallel with any science instrument that is mounted on the telescope.
As a science instrument, the FPI+ is intended to be used as a fast framerate imaging photometer in the visual wavelength range. The highly configurable readout modes of the camera can be adapted to the proposed observation needs. Examples for the scientific use of the FPI+ include observations of stellar occultations and exo-planet transits. The observations of stellar occultations benefit from SOFIA's mobility, e.g. the abilities to fly into the shadow path and to avoid cloud cover. The observation of exo-planet transits benefit from the much reduced scintillation noise at flight altitude, resulting in higher signal-to-noise ratios in the light curves compared to ground based measurements.
Most of the visual light passes SOFIA’s tertiary beam splitter (M3-1) before it is reflected into the Nasmyth tube by the fully-reflective tertiary (M3-2). A significant amount of visual light is not transmitted, but rather absorbed or reflected along with the longer, infrared wavelengths. However, in the range between 480 nm to 800 nm, where the visual-light CCD cameras are most sensitive, more than 50% of the light is transmitted to the FPI+. The visual light continues through a set of four silver-coated folding mirrors inside the so called delay line assembly of the telescope. This setup allows focusing the FPI+ independently from the instrument at the telescope science instrument flange. A pair of windows is installed between the Nasmyth tube and the delay line that create the boundary between the stratospheric conditions in the telescope cavity and cabin conditions inside the delay line assembly. Two eyepiece lenses are used to collimate the telescope beam. Close to the camera is a pellicle beam splitter made of a nitrocellulose membrane with 85% transmission. The beam splitter can be used to reflect a reticle into the light path for camera alignment purposes. The last optical element in front of the camera is an industrial ZEISS 1.4/85 mm Planar T* IR photo lens.
A double-carousel, filter wheel with six positions on each carousel is installed between the reticle beam splitter and the ZEISS lens.
18.104.22.168 Angular Resolution
The CCD sensor of the FPI+ is an e2v CCD201-20 1024 x 1024 pixel frame transfer EMCCD with a plate scale of 0.51 arcsec/pix and a square field of view (FOV) of 8.7 x 8.7 arcmin. The unvignetted FOV is a circular beam of approx. 9 arcmin diameter centered on the FPI+ sensor. Pixel binning of 2x2, 4x4, etc. is available and can be used to increase the frame rate and reduce the effective readout noise. In flight, the seeing blur size of the observatory is at about 4 arcsec diameter. Therefore, a reduction of the angular resolution by binning up to 4x4 (2x2 arcsec2) still provides critical sampling of the seeing element.
The image quality in visible wavelengths on SOFIA is dominated by seeing and image motion effects. A wavelength dependent analysis of the image quality for the visual wavelength range has been done by the HIPO team.
22.214.171.124 Filter Suite
The wavelength range of the FPI+ is 360 nm to 1100 nm. Six spectral filters are available within this range. These are five Sloan Digital Sky Survey filters u’g’r’i’z’ and a Schott RG1000 near-IR cut-on filter. Additionally, three neutral density (ND) filters can be used to attenuate bright stars. The ND filters are required for the tracking function of the FPI+ and the optical densities are chosen in such a way that stars within the brightness range of 0 < V mag < 16 can be imaged with an exposure time of 1 second. The Daylight filter is also a requirement for telescope tracking to be able to acquire bright guide stars in twilight.
|Filter Wheel Configuration|
|Carousel 1||Carousel 2|
|Sloan u'||ND 1|
|Sloan g'||ND 2|
|Sloan r'||ND 3|
Table 6-1 shows the configuration of the FPI+ double filter wheel. Filters from carousel one and two can be combined freely with a few exceptions. The daylight tracking filter from carousel two can only be used with the OPEN position of carousel one to avoid non-overlapping wavelength bands. The Blocked position cannot be selected for observations, but instead is used for taking calibration data (bias frames, dark frames).
Figure 6-1 shows a plot of the FPI+ total system throughput, which includes a model for atmospheric extinction, the calculated SOFIA telescope throughput, and the instrument quantum efficiency. The filter spectral response has been measured and is added to the plot. At the wavelengths where the Sloan u’ filter is transparent, other elements in the FPI+ light path (dichroic tertiary mirror, protected silver coatings, ZEISS lens) are nearly opaque. This results in a very low throughput (~0.5%) for the selection of the FPI+ with the Sloan u’ filter.
Figure 6-2 is a plot of the neutral density filter transmittance vs. wavelength for the three installed ND filters. Over the entire wavelength range of the FPI+, the ND filters have the average optical density listed in Table 6-1. However, there is a wavelength dependence of the optical density of all ND filters that has to be considered when using the ND filters in conjunction with the Sloan filters. All filters are par focal despite their different thicknesses, because they are installed in the parallel beam in front of the Zeiss lens.
|Neutral Density Filter Properties|
|Filter name||Glass Type||Thickness||Average Optical Density|
126.96.36.199 Imaging Sensitivities
The instrument sensitivity and resolution is provided to analyze the feasibility of scientific investigations. The sensitivity of the FPI+ in its different Sloan filters was measured in-flight as part of the camera upgrade verification. The selected star field had targets with a wide range in V mag brightness (11.1 < V mag < 16.7). The full-frame images were acquired with an exposure time of one second without pixel binning. The SNR values in Figure 6-3 are calculated with the measured signal values and the known noise sources. Displayed are the results of the OPEN configuration and the Sloan filters u’, g’, r’, i’, z’ and DAYLIGHT.
188.8.131.52 Camera Performance
|Camera Modes and Performance|
|FPI+ Observing Mode||Horizontal Clock Rate||Bit Depth||Gain [e-/DU]||Minimum Read Noise||FPI+ Tracking Possible|
|FAST_STARE||10 MHz||14 bit||10.7||45.9 e- rms||No|
|FPI_TRACK_MEDIUM_STARE||5 MHz||14 bit||8.9||36.1 e- rms||Yes|
|FPI_TRACK_SLOW_STARE||1 MHz||16 bit||0.7||6.0 e- rms||Yes|
With the camera’s multi-stage thermo-electric cooler, it is possible to achieve sensor temperatures of 100˚ C below ambient temperature. The measured dark current rate at a sensor temperature of -85˚ C, the recommended setting, is 0.001 e-/pixel/second.
The frame rates listed in Table 6-4 are for the full frame. When sub-frames are used (without FPI+ tracking) the achievable rates can be increased. The frame rate then depends on the sub-frame size and its position on the sensor.
|Frame Rates (frames per sec) for the Acquisition of Full Frames|